IEC 60227-1 Cable Testing Equipment Bending Test Apparatus With 60
Per Minute Flexing Rate
IEC60227-1 clause 184.108.40.206 Bending test for tinsel cord:
See 3.2 of IEC 60227-2.
During the test with 60 000 bending cycles, i.e. 120 000 single
strokes, interruption of the current shall not occur. Cable Testing Equipment
After the test, the sample shall withstand the voltage test carried
out in accordance with 2.2 of IEC 60227-2, the voltage, however,
being 1 500 V and applied only between the conductors connected
together and the water.
The cord bending tester is designed and manufactured according to
IEC 60227-1 clause 220.127.116.11, IEC 60227-2 clause 3.2. It is used to
checking the power cord flexing(bending) resistance performance.
Test object of rated voltages up to and including 450/750V
polyvinyl chloride insulated cables. Cable Testing Equipment
The test device is easy to operate, can preset the number of test
times (up to 999999 times), the test speed and test angle can be
adjusted, load current 0-1A stepless adjustable, and with the
current conduction detection function. Tester will stop
automatically when preset test times completed or detected
conduction interrupt, record the tested times and alarm with sound.
The equipment is manufactured according to the standard and the
sample fixture matches the multiple samples. The utility model has
the advantages of beautiful appearance, strong function and high
reliability, and is an indispensable equipment for cable
manufacturers and laboratory testing. Cable Testing Equipment
|Input Power||230V, 50Hz|
|Work station||1 station|
|Flexing angle||Each side 45°, 60°, 90°, has a dial for corresponding reference.
|Flexing speed rate||10~60times per min, manual adjust|
|Test times||0~999999 times, can be preset and displayed|
|Output voltage||12V Cable Testing Equipment|
|Specimen fixture||Specific fixtures|
|Conduction detection||The test device is equipped with power supply terminals for the
test samples, and it has a conduction detection function, if there
is no current go through the test sample, the indicator light turns
on, alarm occurs and the test stops, records the real-time test
Picture for reference: